Liquid Level Instrument Installation Guide (4)
The installation tips of temperature instrument:
If temperature instrument is installed on process pipe with much dust around,it is very necessary to take measures to avoid abrasion.
If thermal resistance or thermocouple is installed in the position where the thermal resistance or thermocouple is easily destroyed by medium, and thermal resistance or thermocouple is inserted into medium in 1 meter depth or the measured temperature is more than 700 celsius, it is very necessary to take measures to avoid bending.
Make sure the sensing surface of temperature sensor and measured medium surface are closely and firmly attached.
Temperature wrap of pressure temperature meter should be in medium completely.There should be protection for accessory of capillary tubes.The bending radius should be within 50mm. Heat insulation measures should be taken if the surrounding temperature fluctuates sharply.